Automatic upper arm blood pressure monitor M2+ Intellisense, 1 piece, Omron
Automatic upper arm blood pressure monitor M2+ Intellisense, 1 piece, OmronProperties: - automatic upper arm blood pressure monitor; - digital LCD display; - blood pressure measurement range: 0 - 299 mmHg; - pulse measurement range: between 40 and 180 beats/minute; - accuracy: Pressure: ±3 mmHg / Pulse: ±5% of the displayed value; - inflation: automatic, with electric pump; - deflation: Automatic deflation valve; - measurement method: oscillometric method; - IP classification: blood pressure monitor: IP20 - power supply: 4 x 1.5 V "AA" batteries or optional AC adapter (AC INPUT 100 - 240 V 50 - 60 Hz 0.12 - 0.065 A); - memory: Stores up to 30 measurement values; - operating conditions: between +10 and +40 °C / between 15 and 90% RH (non-condensing) / between 800 and 1,060 hPa; - storage/transport conditions: between -20 and +60 °C / between 10 and 95% RH (non-condensing); - protection against electric shock: Internally powered ME equipment (when using only batteries) / ME equipment Class II (optional AC adapter); - weight: Blood pressure monitor: approx. 250 g (without batteries); - dimensions (approximate value): Blood pressure monitor: approx. 103 mm (w) × 82 mm (h) × 140 mm (l); How to use: To take a measurement, you should relax and sit comfortably, in a sitting position, with your feet supported. 30 minutes before measuring your blood pressure, do not take a bath, do not consume alcohol or caffeine, do not smoke, do not exercise. Sit with your back straight, without crossing your legs. The arm on which the measurement is made must be supported and the arm cuff must be positioned at the same level as the heart. Do not move or talk during the measurement. Presentation: 1 piece